### Abstract

PURPOSE: To evaluate relative diagnostic precision and test-retest variability of 2 devices, the Compass (CMP, CenterVue, Padova, Italy) fundus perimeter and the Humphrey Field Analyzer (HFA, Zeiss, Dublin, CA), in detecting glaucomatous optic neuropathy (GON).

DESIGN: Multicenter, cross-sectional, case-control study.

PARTICIPANTS: We sequentially enrolled 499 patients with glaucoma and 444 normal subjects to analyze relative precision. A separate group of 44 patients with glaucoma and 54 normal subjects was analyzed to assess test-retest variability.

METHODS: One eye of recruited subjects was tested with the index tests: HFA (Swedish interactive thresholding algorithm [SITA] standard strategy) and CMP (Zippy Estimation by Sequential Testing [ZEST] strategy), 24-2 grid. The reference test for GON was specialist evaluation of fundus photographs or OCT, independent of the visual field (VF). For both devices, linear regression was used to calculate the sensitivity decrease with age in the normal group to compute pointwise total deviation (TD) values and mean deviation (MD). We derived 5% and 1% pointwise normative limits. The MD and the total number of TD values below 5% (TD 5%) or 1% (TD 1%) limits per field were used as classifiers.

MAIN OUTCOME MEASURES: We used partial receiver operating characteristic (pROC) curves and partial area under the curve (pAUC) to compare the diagnostic precision of the devices. Pointwise mean absolute deviation and Bland-Altman plots for the mean sensitivity (MS) were computed to assess test-retest variability.

RESULTS: Retinal sensitivity was generally lower with CMP, with an average mean difference of 1.85±0.06 decibels (dB) (mean ± standard error, P < 0.001) in healthy subjects and 1.46±0.05 dB (mean ± standard error, P < 0.001) in patients with glaucoma. Both devices showed similar discriminative power. The MD metric had marginally better discrimination with CMP (pAUC difference ± standard error, 0.019±0.009, P = 0.035). The 95% limits of agreement for the MS were reduced by 13% in CMP compared with HFA in participants with glaucoma and by 49% in normal participants. Mean absolute deviation was similar, with no significant differences.

CONCLUSIONS: Relative diagnostic precision of the 2 devices is equivalent. Test-retest variability of MS for CMP was better than for HFA.

Original language | English |
---|---|

Article number | S0161-6420(18)30935-7 |

Journal | Ophthalmology |

DOIs | |

Publication status | E-pub ahead of print - Aug 14 2018 |

### Fingerprint

### Cite this

*Ophthalmology*, [S0161-6420(18)30935-7]. https://doi.org/10.1016/j.ophtha.2018.08.010

**A Comparison between the Compass Fundus Perimeter and the Humphrey Field Analyzer.** / Montesano, Giovanni; Bryan, Susan R; Crabb, David P; Fogagnolo, Paolo; Oddone, Francesco; McKendrick, Allison M; Turpin, Andrew; Lanzetta, Paolo; Perdicchi, Andrea; Johnson, Chris A; Garway-Heath, David F; Brusini, Paolo; Rossetti, Luca M.

Research output: Contribution to journal › Article

*Ophthalmology*. https://doi.org/10.1016/j.ophtha.2018.08.010

}

TY - JOUR

T1 - A Comparison between the Compass Fundus Perimeter and the Humphrey Field Analyzer

AU - Montesano, Giovanni

AU - Bryan, Susan R

AU - Crabb, David P

AU - Fogagnolo, Paolo

AU - Oddone, Francesco

AU - McKendrick, Allison M

AU - Turpin, Andrew

AU - Lanzetta, Paolo

AU - Perdicchi, Andrea

AU - Johnson, Chris A

AU - Garway-Heath, David F

AU - Brusini, Paolo

AU - Rossetti, Luca M

N1 - Copyright © 2018 American Academy of Ophthalmology. Published by Elsevier Inc. All rights reserved.

PY - 2018/8/14

Y1 - 2018/8/14

N2 - PURPOSE: To evaluate relative diagnostic precision and test-retest variability of 2 devices, the Compass (CMP, CenterVue, Padova, Italy) fundus perimeter and the Humphrey Field Analyzer (HFA, Zeiss, Dublin, CA), in detecting glaucomatous optic neuropathy (GON).DESIGN: Multicenter, cross-sectional, case-control study.PARTICIPANTS: We sequentially enrolled 499 patients with glaucoma and 444 normal subjects to analyze relative precision. A separate group of 44 patients with glaucoma and 54 normal subjects was analyzed to assess test-retest variability.METHODS: One eye of recruited subjects was tested with the index tests: HFA (Swedish interactive thresholding algorithm [SITA] standard strategy) and CMP (Zippy Estimation by Sequential Testing [ZEST] strategy), 24-2 grid. The reference test for GON was specialist evaluation of fundus photographs or OCT, independent of the visual field (VF). For both devices, linear regression was used to calculate the sensitivity decrease with age in the normal group to compute pointwise total deviation (TD) values and mean deviation (MD). We derived 5% and 1% pointwise normative limits. The MD and the total number of TD values below 5% (TD 5%) or 1% (TD 1%) limits per field were used as classifiers.MAIN OUTCOME MEASURES: We used partial receiver operating characteristic (pROC) curves and partial area under the curve (pAUC) to compare the diagnostic precision of the devices. Pointwise mean absolute deviation and Bland-Altman plots for the mean sensitivity (MS) were computed to assess test-retest variability.RESULTS: Retinal sensitivity was generally lower with CMP, with an average mean difference of 1.85±0.06 decibels (dB) (mean ± standard error, P < 0.001) in healthy subjects and 1.46±0.05 dB (mean ± standard error, P < 0.001) in patients with glaucoma. Both devices showed similar discriminative power. The MD metric had marginally better discrimination with CMP (pAUC difference ± standard error, 0.019±0.009, P = 0.035). The 95% limits of agreement for the MS were reduced by 13% in CMP compared with HFA in participants with glaucoma and by 49% in normal participants. Mean absolute deviation was similar, with no significant differences.CONCLUSIONS: Relative diagnostic precision of the 2 devices is equivalent. Test-retest variability of MS for CMP was better than for HFA.

AB - PURPOSE: To evaluate relative diagnostic precision and test-retest variability of 2 devices, the Compass (CMP, CenterVue, Padova, Italy) fundus perimeter and the Humphrey Field Analyzer (HFA, Zeiss, Dublin, CA), in detecting glaucomatous optic neuropathy (GON).DESIGN: Multicenter, cross-sectional, case-control study.PARTICIPANTS: We sequentially enrolled 499 patients with glaucoma and 444 normal subjects to analyze relative precision. A separate group of 44 patients with glaucoma and 54 normal subjects was analyzed to assess test-retest variability.METHODS: One eye of recruited subjects was tested with the index tests: HFA (Swedish interactive thresholding algorithm [SITA] standard strategy) and CMP (Zippy Estimation by Sequential Testing [ZEST] strategy), 24-2 grid. The reference test for GON was specialist evaluation of fundus photographs or OCT, independent of the visual field (VF). For both devices, linear regression was used to calculate the sensitivity decrease with age in the normal group to compute pointwise total deviation (TD) values and mean deviation (MD). We derived 5% and 1% pointwise normative limits. The MD and the total number of TD values below 5% (TD 5%) or 1% (TD 1%) limits per field were used as classifiers.MAIN OUTCOME MEASURES: We used partial receiver operating characteristic (pROC) curves and partial area under the curve (pAUC) to compare the diagnostic precision of the devices. Pointwise mean absolute deviation and Bland-Altman plots for the mean sensitivity (MS) were computed to assess test-retest variability.RESULTS: Retinal sensitivity was generally lower with CMP, with an average mean difference of 1.85±0.06 decibels (dB) (mean ± standard error, P < 0.001) in healthy subjects and 1.46±0.05 dB (mean ± standard error, P < 0.001) in patients with glaucoma. Both devices showed similar discriminative power. The MD metric had marginally better discrimination with CMP (pAUC difference ± standard error, 0.019±0.009, P = 0.035). The 95% limits of agreement for the MS were reduced by 13% in CMP compared with HFA in participants with glaucoma and by 49% in normal participants. Mean absolute deviation was similar, with no significant differences.CONCLUSIONS: Relative diagnostic precision of the 2 devices is equivalent. Test-retest variability of MS for CMP was better than for HFA.

U2 - 10.1016/j.ophtha.2018.08.010

DO - 10.1016/j.ophtha.2018.08.010

M3 - Article

JO - Ophthalmology

JF - Ophthalmology

SN - 0161-6420

M1 - S0161-6420(18)30935-7

ER -